Nanopia TDM Valproic Acid
We created this page for medical staff.
*Read carefully the package insert before use.
- Liquid reagent that requires no reagent preparation
- Applicable to various types of automated analyzers
- *This package insert follows the Pharmaceuticals, Medical devices and Other Therapeutic Products Act of Japan.
Latex-enhanced turbidimetric immunoassay
When a certain amount of anti- valproic acid antibody is added to a specimen, the anti- valproic acid antibody binds to valproic acid until it has reacted with all of the available valproic acid in the specimen. Next, valproic acid -coated latex are added, which react with the residual anti- valproic acid antibody, leading to agglutination. Since the extent of agglutination depends on the concentration of valproic acid in the specimen, the valproic acid concentration can be determined by measuring the absorbance as an indicator of agglutination.
|Nanopia TDM Valproic Acid||VPA Antibody Solution 1||1×30mL||2-8℃|
|PHT Latex Reagent 2||1×10mL|
- *Other package sizes are available. Contact SEKISUI MEDICAL CO., LTD. for details.
|TDM Calibrator for Nanopia **||1×6conc.×2mL||2-8℃|
|TDM Sample Dilution Solution for Nanopia **||1×10mL|
- **Not in vitro diagnostic products in Japan.
"Nanopia" is a trademark or registered trademark of SEKISUI MEDICAL CO., LTD. in Japan and/or other countries.
TDM (Therapeutic Drug Monitoring) Product List
We prepared this page for medical staff (doctors, dentists, pharmacists, clinical laboratory technicians, nurses, etc.) in Japan to ensure the correct use of our products. You should be aware that the information provided is not intended for overseas medical staff and the general public.